This content will become publicly available on June 1, 2023

A cryogenic torsion balance using a liquid-cryogen free, ultra-low vibration cryostat
We describe a liquid-cryogen free cryostat with ultra-low vibration levels, which allows for continuous operation of a torsion balance at cryogenic temperatures. The apparatus uses a commercially available two-stage pulse-tube cooler and passive vibration isolation. The torsion balance exhibits torque noise levels lower than room temperature thermal noise by a factor of about four in the frequency range of 3–10 mHz, limited by residual seismic motion and by radiative heating of the pendulum body. In addition to lowering thermal noise below room-temperature limits, the low-temperature environment enables novel torsion balance experiments. Currently, the maximum duration of a continuous measurement run is limited by accumulation of cryogenic surface contamination on the optical elements inside the cryostat.
Authors:
; ; ; ; ; ; ;
Award ID(s):
Publication Date:
NSF-PAR ID:
10339931
Journal Name:
Review of Scientific Instruments
Volume:
93
Issue:
6
Page Range or eLocation-ID:
064505
ISSN:
0034-6748
5. Controlling stray light at millimeter wavelengths requires special optical design and selection of absorptive materials that should be compatible with cryogenic operating environments. While a wide selection of absorptive materials exists, these typically exhibit high indices of refraction and reflect/scatter a significant fraction of light before absorption. For many lower index materials such as commercial microwave absorbers, their applications in cryogenic environments are challenging. In this paper, we present a new tool to control stray light: metamaterial microwave absorber tiles. These tiles comprise an outer metamaterial layer that approximates a lossy gradient index anti-reflection coating. They are fabricated via injection molding commercially available carbon-loaded polyurethane (25% by mass). The injection molding technology enables mass production at low cost. The design of these tiles is presented, along with thermal tests to 1 K. Room temperature optical measurements verify their control of reflectance to less than 1% up to$65∘<#comment/>$angles of incidence, and control of wide angle scattering below 0.01%. The dielectric properties of the bulk carbon-loaded material used in the tiles is also measured at different temperatures, confirming that the material maintains similar dielectric properties down to 3 K.